The Cervico Premium Kit includes both Cervico Mold & Cervico Guide, for a discounted price.
Cervico by VPI General Information and Instructions of use
The product is available in 3 different versions which are encoded as
vs. Cervico mold (Item No #5213006650000)
includes only the Alum mold and Silicone insert plus an allen head screwdriver which is used for the positioning of the inserts and for removing the silicon insert from the alum base.
vs. Cervico Guide (Item No #5213006650017)
includes only the guide and its components (1 silicone autoclavable guide tray, 20 tabs, 1 guide retention handle, 4 Guide pins).
vs. Cervico Premium (Item No #5213006650024)
includes both the guide and the mold with their aforementioned components.
Use The guide of the system (Cervico Guide) is used for the intraoral and / or extraoral evaluation of a patient's edentulous space in order to choose the appropriate size and shape customized abutment and duplicate impression post to be manufactured from the mold of the system. In addition, the tabs of the guide and / or the pins of the guide, individually or in combination, may be used during the osteotomy preparation in the patient's jaw for the purpose of marking the correct position of the osteotomy and controlling the inclination of the latter during the steps of bone drilling. All functional parts of the guide can be re-used after proper disinfection and sterilization process following the same standards as for sterilization process of reusable dental tools.
The mold of the system (Cervico Mold) is used for the manufacturing of customized abutments and customised impression posts. All functional parts of the mold (base, silicone insert) can be re-used after proper disinfection and a sterilization process following the same standards as for sterilization process of reusable dental tools.
Each of the tabs of the guide has a shape and dimensions that are the same to the cervical portion of a corresponding well of the mold that carries the same coding as the one of the tab to which it corresponds. E.g. If the tab (pS) of the guide has been chosen as the one representing best the cervical profile of an implant site, then the (pS) well of the mold must be used in order to produce the custom healing abutment and it's duplicate impression post to be utilised for this case.